08/03/22 - Weather station set up in Corton Denham
09/03/22 - Wind direction stops working, stuck on North.
Contacted Weather shop for support
Response from Davis via Weather Shop to change station ID - done but no improvement.
10/03/22 - Awaiting response from Weather Shop / Davis
13/03/22 - New Weather Underground ID obtained, ISHERB47 - now live on WU
16/03/22 - Basic website launched, more features to be added gradually
17/03/22 - Event log added to website
17/03/22 - Response from Weather Shop, they are sending anemometer and wind vane test cable.
23/03/22 - Test cable received
24/03/22 - Test performed. This has confirmed that the fault lies either in the anemometer head or in the
connecting cable. The cable has no visible damage, but since it is permanently connected to the
anemometer, I can't do anything else without invalidating the warranty. In the warmer weather I have
noticed that the wind vane starts working (intermittently) as the temperature rises.
28/03/22 - New anemometer received from the Weather Shop, who have given me good support. Fitted and it now
- indicates wind direction correctly. Hopefully that's the end of this saga!